|Year : 2019 | Volume
| Issue : 4 | Page : 241-248
Adverse events with lumen-apposing metal stents in endoscopic gallbladder drainage: A systematic review and meta-analysis
Babu P Mohan1, Ravishankar Asokkumar2, Mohammed Shakhatreh3, Rajat Garg4, Suresh Ponnada5, Udayakumar Navaneethan6, Douglas G Adler7
1 Inpatient Medicine, Banner University Medical Center/ University of Arizona, Tucson, Arizona, USA
2 Division of Gastroenterology and Hepatology, Singapore General Hospital, Singapore
3 Department of Hospital Medicine, Rapides Regional Medical Center, Alexandria, Louisiana, USA
4 Department of Hospital Medicine, Cleveland Clinic Foundation, Cleveland, Ohio, USA
5 Department of Hospital Medicine, Carilion Roanoke Memorial Hospital, Roanoke, Virginia, USA
6 Center for Interventional Endoscopy, Florida Hospital, Orlando, Florida, USA
7 Division of Gastroenterology and Hepatology, University of Utah School of Medicine, Salt Lake City, Utah, USA
|Date of Submission||10-Oct-2018|
|Date of Acceptance||15-Dec-2018|
|Date of Web Publication||09-May-2019|
Douglas G Adler
Division of Gastroenterology and Hepatology, Huntsman Cancer Center, University of Utah School of Medicine, 30N 1900E 4R118, Salt Lake City, Utah 84132
Source of Support: None, Conflict of Interest: None
Background: Lumen-apposing metal stents (LAMS) are rapidly being used in endoscopic interventional drainage procedures and have started to replace the self-expanding metal stents (SEMSs). Its use in gallbladder drainage (GBD) is limited by lack of good-quality studies, and data are scarce on its safety. Methods: We conducted a comprehensive search of multiple electronic databases and conference proceedings including PubMed, Embase, and Web of Science databases (from inception through July 2018) to identify studies that reported on the use of LAMS in GBD. The outcomes measured were the pooled rates of all adverse events (AEs), pooled rates of early AEs and pooled rates of delayed AEs. Results: A total of 8 studies (393 patients) were included. The pooled rate of all AEs was 12.7% (95% CI 8.4-18.7, I2= 7.7) compared to 17.5% (95% CI 10.2-28.2, I2= 65.1) with other SEMS, P = 0.39. The rate of early AEs with LAMS in endoscopic ultrasound-GBD (EUS-GBD) was 6.5% (95% CI 4.2-10, I2= 1.2), and the rate of delayed AEs was 8.3% (95% CI 5.8-11.9, I2 = 4.8). The rate of recurrent cholecystitis and/or cholangitis was 4.6% (95% CI 2.6-9.5, I2= 0) and the pooled rate of death was 5% (95% CI 2.6-9.5, I2 = 36.4). Conclusion: We report an overall AE rate of 13% with LAMS in EUS-GBD. Early AE risk appears to be 6.5% and delayed AE risk appears to be 8%. Our results are analyzed out of good-quality studies, with minimal to zero heterogeneity.
Keywords: Adverse events, gallbladder drainage, lumen-apposing metal stents, self-expanding metal stents
|How to cite this article:|
Mohan BP, Asokkumar R, Shakhatreh M, Garg R, Ponnada S, Navaneethan U, Adler DG. Adverse events with lumen-apposing metal stents in endoscopic gallbladder drainage: A systematic review and meta-analysis. Endosc Ultrasound 2019;8:241-8
|How to cite this URL:|
Mohan BP, Asokkumar R, Shakhatreh M, Garg R, Ponnada S, Navaneethan U, Adler DG. Adverse events with lumen-apposing metal stents in endoscopic gallbladder drainage: A systematic review and meta-analysis. Endosc Ultrasound [serial online] 2019 [cited 2019 Aug 19];8:241-8. Available from: http://www.eusjournal.com/text.asp?2019/8/4/241/257857
| Introduction|| |
Lumen-apposing metal stents (LAMS) were first introduced in 2012 and made widely clinically available in 2015. The different LAMS commercially available for use are the AXIOS stent (by Xlumena Inc., Mountain View, CA, USA); the Niti-S SPAXUS and NAGI stents (by Taewoong-Medical Co., Ilsan, Korea); the AIX stent (by Leufen Medical, Aachen, Germany); and the brachiocephalic fistula (BCF) stent (by Hanaro MRI Tech). All are made of nitinol wire and fully covered with silicon. The shape of a LAMS is designed to keep the target object (pseudocyst, walled-off necrosis, etc.) closely apposed to the gastrointestinal (GI) lumen. They are now widely employed in many clinical settings, including the treatment of benign luminal GI strictures, endoscopic drainage of pancreatic fluid collections, endoscopic gallbladder drainage (E-GBD), and endoscopic gastrojejunostomy.
Like any other advanced endoscopic intervention, the use of LAMS carries risks of adverse events (AEs). Early recognition of AEs is critical to minimize morbidity and mortality. These AEs can be immediate during the stent deployment or delayed. Data are limited regarding the safety of LAMS when used to provide E-GBD. The reported results are highly varied as the current literature consists of many small retrospective case series. Recently published qualitative and quantitative reviews are limited by inclusion of these small studies. Small and low-quality studies add to the heterogeneity and negatively impact the overall reliability of the results. Despite the limited data on the safety of LAMS, there has been clinical adoption of this stent globally in setting where a LAMS is used, and studies have suggested that it can be safely used in an outpatient setting.
We conducted this meta-analysis, using only good-quality studies, to better understand and quantify the AE rates with the use of LAMS in GBD.
| Methods|| |
We conducted a comprehensive search of several databases and conference proceedings including PubMed, Embase, and Web of Science databases (earliest inception to July 2018). We followed the preferred reporting items for systematic reviews and meta-analyses guidelines, by using predefined protocol, to identify studies reporting the use of LAMS in GBD. An experienced medical librarian using inputs from the study authors helped with the literature search.
Keywords used in the search included a combination of ≴lumen-apposing metal stent,≵ ≴AXIOS stent,≵ ≴SPAXUS stent,≵ ≴NAGI stent,≵ ≴NITI-S stent,≵ ≴AIX stent,≵ ≴BCF stent,≵ ≴endoscopic ultrasound guided,≵ and ≴gall-bladder drainage.≵ The search was restricted to studies in human subjects and published in English language in peer-reviewed journals. Three authors (M.B., S.M., A.R.) independently reviewed the title and abstract of studies identified in primary search and excluded studies that did not address the research question, based on prespecified exclusion and inclusion criteria. The full text of remaining articles was reviewed to determine whether it contained relevant information. Any discrepancy in article selection was resolved by consensus and in discussion with a coauthor.
The bibliographic section of the selected articles as well as the systematic and narrative articles on the topic was manually searched for additional relevant articles.
In this meta-analysis, we included cohort studies that met the following criteria: (1) use of LAMS in GBD and (2) data on AEs attributed to LAMS. Studies irrespective of the reason for GBD, the size of LAMS, the commercial type of LAMS, geography, and abstract/manuscript status were included as long as they provided data needed for the analysis.
Following were our exclusion criteria: (1) studies that provided no data on AEs attributed to LAMS, (2) studies that reported on self-expanding metal stents (SEMSs) that were not lumen apposing, and (3) <30 patients in the study cohort.
In case of multiple publications from the same cohort, data from the most recent and/or most appropriate comprehensive report were included. In our search process, we encountered two such studies, one by Jang et al. and other by Irani et al.
Data abstraction and quality assessment
Data on study-related outcomes in the individual studies were abstracted onto a standardized form by at least 2 authors (M. B., S. M.) independently, and 3 authors (M. B., S. M., G. R.) did the quality scoring independently.
Using a scale modified from the Newcastle–Ottawa scale for cohort studies assessed the quality of included studies. This quality score consisted of 6 questions: representative of the average adult in the community (1 point for population-based studies, 0.5 point for multicenter studies; 0 point for a single-center hospital-based study); large cohort size (1 point if >30 patients, 0.5 point if between 15 and 30 patients, 0 point if <15 patients); information on overall AEs (1 point if reported; 0 point if not reported); information on subtypes of AEs (1 point if reported, 0 point if not reported); type of article write-up (1 point if original manuscript, 0.5 point if abstract); and attrition rate (1 point if all patients were accounted for, 0.5 point if <50% patients lost to follow-up, 0 point if >50% patients lost to follow-up). A score of ≥5, 3–4, and <3 was considered suggestive of high-quality, medium-quality, and low-quality study, respectively.
The primary analysis focused on assessing the pooled rate of overall AEs in EUS-GBD with LAMS. The pooled rate of AE in EUS-GBD with other SEMS was used as a comparator.
Our secondary analysis focused on calculating the following in EUS-GBD with LAMS:
- Pooled rates of early AE; with further classification into bleeding, bile leak, stent occlusion, perforation, and stent migration
- Pooled rates of delayed AE; with further classification of the delayed AEs into bleeding, bile leak, stent occlusion, and stent migration
- Pooled rates of recurrent cholecystitis and/or cholangitis and death, and
- Pooled rates of technical success and clinical success.
Majority of studies defined AEs as any procedure, drain or stent-related event. Studies did not mention any predetermined criterion to define bile leak, perforation, stent occlusion, and stent migration. These were considered as AEs even if the patient suffered no clinical symptoms out of the event. Bleeding was defined as significant if requiring blood product transfusion, endoscopic therapy, and radiologic or surgical interventions. AEs reported within the first 2 weeks of EUS-guided LAMS placement were considered early, and the ones reported after 2 weeks of intervention were considered delayed.
Technical success was defined as the ability to place a transmural LAMS into the gallbladder and/or the bile duct as determined by the flow of bile and/or pus. Clinical success was defined as the resolution of symptoms, laboratory abnormalities, and/or radiologic abnormalities. There was variability in the laboratory resolution cutoff among studies, with majority defining it as a decrease in bilirubin levels to <10% of initial levels and/or <50% of initial levels with resolution of clinical symptoms.
We used meta-analysis techniques to calculate the pooled estimates in each case following the methods suggested by DerSimonian and Laird using the random effects model. When the incidence of an outcome was zero in a study, a correction of 0.01 was added to the number of incident cases before statistical analysis. We assessed heterogeneity between study-specific estimates using two methods, the Cochran Q statistical test for heterogeneity and the I2 statistic., In this, values of <30%, 30%–60%, 61%–75%, and >75% were suggestive of low, moderate, substantial, and considerable heterogeneity, respectively. Since random effects model estimates an average effect, we also calculated the 95% prediction interval, which deals with the dispersion of the effects., Publication bias was ascertained, qualitatively, by visual inspection of funnel plot, and quantitatively, by the Egger test.
All analyses were performed using comprehensive meta-analysis software, version 3 (BioStat, Englewood, NJ, USA).
| Results|| |
Search results and population characteristics
From an initial total of 658 citations identified using our search strategy, our search of the database resulted in 36 studies that met our inclusion criteria. Of these, 18 studies were excluded because they had <30 patients in their study cohort and 10 studies were excluded because of other methods and/or routes used to drain the gallbladder. A total of 8 studies,,,,,,, were included in the final analysis. All studies used the AXIOS stent except one study that used the Spaxus stent. The schematic diagram of study selection is illustrated in [Figure 1]. [Table 1] describes the study and population characteristics.
Majority of the population consisted of males (37%–65%), with an age range of 25–93 years. Procedure time was reported in the range of 8–104 min. Acute cholecystitis and/or malignancy was the clinical indication for EUS-GBD. Period of follow-up ranged from 1 day to 411 days.
Characteristics and quality of included studies
[Table 1] describes the study and population characteristics. None of the studies were population based. Six studies,,,,, were multicenter based and the rest 2, were from single center. All studies had more than or equal to 30 patients. All studies reported on the overall AEs but were variable in the subtypes. One study by Garcia-Alonso et al. reported only on stent migration and gastrointestinal bleeding. All studies except two by Anderloni et al. and Teoh et al., were full manuscripts and all studies had adequate follow up of their study population. Overall, 6 studies,,,,, were considered to be of high quality and the rest 2, were of medium quality. None were of low quality. The study by Walter et al. was a prospectively designed study [Supplement Table 1 [Additional file 1]] Study quality assessment].
Adverse events in EUS-gallbladder drainage with lumen-apposing metal stents
The rate of all AEs with LAMS in EUS-GBD (8 studies,,,,,,,, 393 patients, 47 events) was 12.7% (95% CI 8.4-18.7, I2= 7.7), as compared to 17.5% (95% CI 10.2-28.2, I2= 65.1) with the use of other SEMS (6 studies,,,,,, 154 patients), without any statistical significance to the difference, P = 0.39 [Figure 2].
Rates of AE subtypes are as given below:
Procedure and/or lumen-apposing metal stents-related early adverse events
The rate of all early AE with LAMS in EUS-GBD (8 studies,,,,,,,, 393 patients, 19 events) was 6.5% (95% CI 4.2-10, I2 = 1.2) [Figure 3].
Rates of early AE subtypes are as given below:
- Bleeding: 2.6% (95% CI 0.9-7.2, I2 = 47.1)
- Bile leak: 1.3% (95% CI 0.5-3.3, I2 = 0)
- Stent occlusion: No events
- Perforation: 2.3% (95% CI 1.1-4.7, I2 = 0)
- Stent migration: 1.5% (95% CI 0.6-3.5, I2 =0). [forest plots: [Supplement Figure 2 [Additional file 7]], [Supplement Figure 5 [Additional file 8]], [Supplement Figure 8 [Additional file 9]], [Supplement Figure 10 [Additional file 10]].
Lumen-apposing metal stents-related delayed adverse events; recurrent cholecystitis and/or cholangitis; death
The rate of all delayed AE with LAMS in EUS-GBD (8 studies,,,,,,,, 393 patients, 28 events) was 8.3% (95% CI 5.8-11.9, I2 = 4.8) [Figure 4].
Rates of delayed AE subtypes are as given below:
The rate of recurrent cholecystitis and/or cholangitis (6 studies,,,,,, 301 patients, 11 events) was 4.6% (95% CI 2.6-8.0, I2 = 0) [Supplement Figure 12 [Additional file 15]].
The rate of death (7 studies,,,,,,, 337 patients, 15 events) was 5% (95% CI 2.6-9.5, I2 = 36.4) [Supplement Figure 13 [Additional file 16]]. All adverse events reported are summarized in [Table 2] and all pooled results are summarized in [Table 3]. Forest plots for all adverse event sub-types are given in supplementary material.
Technical success and clinical success
The rates of technical success and clinical success with LAMS in EUS-GBD (8 studies,,,,,,,, 393 patients) were 94.9% (95% CI 90.8–97.3, I2 = 25.5) and 94.6% (95% CI 91.4-96.7, I2 =0), respectively [forest plots: [Supplement Figure 14 [Additional file 17]] and [Supplement Figure 15 [Additional file 18]].
Validation of meta-analysis results
To assess whether any one study had a dominant effect on the meta-analysis, we excluded one study at a time and analyzed its effect on the main summary estimate. On this analysis, no single study significantly affected the outcome or the heterogeneity.
Based on Q statistics, and I2 analysis for heterogeneity, low (I2< 30) to no (I2 = 0) heterogeneity was noted in the analyses of the primary and secondary outcomes of LAMS in EUS-GBD.
Based on visual inspection of the funnel plot as well as quantitative measurement that used the Egger regression test, there was no evidence of publication bias [Supplement Figure 16] - funnel plot].
| Discussion|| |
Our meta-analysis is the first study to report pooled data from good-quality studies, avoiding small case studies and reports, on the AEs of LAMS in GBD. We report an overall AEs rate of 13% with LAMS as compared to 17.5% with other SEMS, with no statistical significance. The early AE rate was 6.5%, and the delayed AE rate was 8%.
The calculated early versus delayed rates for bleeding (2.6% vs 2%), bile leak (1.3% vs 1.9%), and stent migration (1.5% vs 3%) were comparable. The calculated risk of perforation was 2.3% that was reported only as an early AE. The calculated rate of stent occlusion was 5.2% that was reported only as a delayed AE. The rate of cholecystitis and/or cholangitis was 4.6%, and the rate of death was 5%. Our calculated rate of death needs to be interpreted with caution. Majority of studies reported death as the end outcome and were mostly related to the underlying disease process. None of the reported death events were related to the procedure and/or the use of LAMS. The calculated technical success was 94.9% and clinical success was 94.6%.
Why do this study and how do our results compare with other recently published similar studies? At the time this study was generated, at least four other systematic reviews were published on similar topic. These are studies by Han et al., Manta et al., Kalva et al., and Jain et al. All studies, including our study, were consistent in the reporting of calculated technical success (93.86%–95%) and clinical success (92.48%–93%) rates. Nevertheless, our study differs significantly from these studies in the calculation and reporting of the AEs. The study by Jain et al. was a qualitative review without a quantitative pooling of results. Manta et al. did not report the pooled rates of AEs. Kalva et al. included the BONA-AL stent (Standard Sci Tech Inc., Seoul, Korea) in their analysis, which technically does not generate lumen-apposing force and hence is not considered a LAMS. Han et al. studied the use of LAMS in pancreatic fluid collections and GBD. They reported a pooled AE rate of 17.1% with LAMS in EUS-GBD without any mention of heterogeneity and/or validation statistics for their meta-analysis. All these studies were severely limited by the inclusion of multiple small studies.
The strengths of our review are as follows: systematic literature search with well-defined inclusion criteria, carefully excluding small studies, case reports and redundant studies, inclusion of high quality studies, detailed extraction of clinically useful data in terms of early and delayed AEs attributed to LAMS, rigorous evaluation of study quality, overall low to zero heterogeneity, and statistics to establish and/or refute the validity of the results of our meta-analysis.
There were few limitations to this study and are the ones expected with any meta-analysis. The included studies were not entirely representative of the general population and community practice. The studies did not uniformly follow the severity of AEs grading according to the lexicon of endoscopic AEs. The data itself were derived from a patient group with high morbidity and/or mortality and were not able to stratify our results based on patient performance characteristics. Paucity of data limited our ability to analyze the events with respect to the commercial type of stent used. More large-scale studies are needed to better compare the safety profiles of the different types of LAMS, along with cost-effectiveness. However, this estimate is still the best available estimate that may be used in counseling patients undergoing GBD with LAMS.
| Conclusion|| |
In conclusion, our estimated risk of LAMS-related AE with EUS-GBD is 13%, which is the lowest reported in literature thus far. The rate of early AE is 6.5%, and the risk of delayed AE is 8%. There was low to no heterogeneity in our analysis.
Supplementary information is linked to the online version of the paper on the Endoscopic Ultrasound website.
Financial support and sponsorship
Conflicts of interest
There are no conflicts of interest.
| References|| |
Itoi T, Binmoeller KF, Shah J, et al.
Clinical evaluation of a novel lumen-apposing metal stent for endosonography-guided pancreatic pseudocyst and gallbladder drainage (with videos). Gastrointest Endosc
Teoh AY, Ng EK, Chan SM, et al. Ex vivo
comparison of the lumen-apposing properties of EUS-specific stents (with video). Gastrointest Endosc
Yang D, Nieto JM, Siddiqui A, et al.
Lumen-apposing covered self-expandable metal stents for short benign gastrointestinal strictures: A multicenter study. Endoscopy
Siddiqui AA, Kowalski TE, Loren DE, et al.
Fully covered self-expanding metal stents versus lumen-apposing fully covered self-expanding metal stent versus plastic stents for endoscopic drainage of pancreatic walled-off necrosis: Clinical outcomes and success. Gastrointest Endosc
Dollhopf M, Larghi A, Will U, et al.
EUS-guided gallbladder drainage in patients with acute cholecystitis and high surgical risk using an electrocautery-enhanced lumen-apposing metal stent device. Gastrointest Endosc
Khashab MA, Bukhari M, Baron TH, et al.
International multicenter comparative trial of endoscopic ultrasonography-guided gastroenterostomy versus surgical gastrojejunostomy for the treatment of malignant gastric outlet obstruction. Endosc Int Open
IntHout J, Ioannidis JP, Borm GF, et al
. Small studies are more heterogeneous than large ones: A meta-meta-analysis. J Clin Epidemiol
Adler DG, Shah J, Nieto J, Binmoeller K, Bhat Y, Taylor LJ, Siddiqui AA. Placement of lumen-apposing metal stents to drain pseudocysts and walled-off pancreatic necrosis can be safely performed on an outpatient basis: A multicenter study. Endosc Ultrasound
[Epub ahead of print]. Available from: http://www.eusjournal.com/preprintarticle.asp?id=232608
. [Last cited on 2019 Jan 02].
Moher D, Liberati A, Tetzlaff J, et al
. Preferred reporting items for systematic reviews and meta-analyses: The PRISMA statement. Ann Intern Med
Jang JW, Lee SS, Song TJ, et al.
Endoscopic ultrasound-guided transmural and percutaneous transhepatic gallbladder drainage are comparable for acute cholecystitis. Gastroenterology
Irani S, Ngamruengphong S, Teoh A, et al.
Similar efficacies of endoscopic ultrasound gallbladder drainage with a lumen-apposing metal stent versus percutaneous transhepatic gallbladder drainage for acute cholecystitis. Clin Gastroenterol Hepatol
Stang A. Critical evaluation of the newcastle-ottawa scale for the assessment of the quality of nonrandomized studies in meta-analyses. Eur J Epidemiol
DerSimonian R, Laird N. Meta-analysis in clinical trials. Control Clin Trials
Sutton AJ, Abrams KR, Jones DR, et al
. Methods for Meta-Analysis in Medical Research. John Wiley & Sons Ltd. New York. 2000; 205-28.
Kanwal F, White D. ≴Systematic reviews and meta-analyses≵ in clinical gastroenterology and hepatology. Clin Gastroenterol Hepatol
Higgins JP, Thompson SG, Deeks JJ, et al
. Measuring inconsistency in meta-analyses. BMJ 2003;327:557-60.
Guyatt GH, Oxman AD, Kunz R, et al.
GRADE guidelines: 7. Rating the quality of evidence – Inconsistency. J Clin Epidemiol
Higgins JP, Thompson SG, Spiegelhalter DJ. A re-evaluation of random-effects meta-analysis. J R Stat Soc Ser A Stat Soc
Riley RD, Higgins JP, Deeks JJ. Interpretation of random effects meta-analyses. BMJ
Easterbrook PJ, Berlin JA, Gopalan R, et al
. Publication bias in clinical research. Lancet
Anderloni AA, Fugazza A, Auriemma F, et al.
Su1416 EUS-guided choledochoduodenostomy using a lumen apposing metal stent for malignant distal biliary obstruction: A restrospective analysis of a single center experience. Gastrointest Endosc
Garcia-Alonso FJ, Sanchez-Ocana R, Peñas-Herrero I, et al.
Cumulative risks of stent migration and gastrointestinal bleeding in patients with lumen-apposing metal stents. Endoscopy
Kunda R, Pérez-Miranda M, Will U, et al.
EUS-guided choledochoduodenostomy for malignant distal biliary obstruction using a lumen-apposing fully covered metal stent after failed ERCP. Surg Endosc
Walter D, Teoh AY, Itoi T, et al.
EUS-guided gall bladder drainage with a lumen-apposing metal stent: A prospective long-term evaluation. Gut
Teoh AY, Chan SM, Rerknimitr R, et al
. Su1386 the use of a newly developed self-approximating lumen apposing metallic stent for drainage of the gallbladder and common bile duct – A multi-centered prospective study. Gastrointestinal Endoscopy
Teoh AY, Serna C, Penas I, et al.
Endoscopic ultrasound-guided gallbladder drainage reduces adverse events compared with percutaneous cholecystostomy in patients who are unfit for cholecystectomy. Endoscopy
Ahmed O, Ogura T, Eldahrouty A, et al.
Endoscopic ultrasound-guided gallbladder drainage: Results of long-term follow-up. Saudi J Gastroenterol
Bang JY, Navaneethan U, Hasan M, et al
. Stent placement by EUS or ERCP for primary biliary decompression in pancreatic cancer: A randomized trial (with videos). Gastrointest Endosc
Choi JH, Lee SS, Choi JH, et al.
Long-term outcomes after endoscopic ultrasonography-guided gallbladder drainage for acute cholecystitis. Endoscopy
Choi JH, Kim HW, Lee JC, et al.
Percutaneous transhepatic versus EUS-guided gallbladder drainage for malignant cystic duct obstruction. Gastrointest Endosc
Song TJ, Hyun YS, Lee SS, et al.
Endoscopic ultrasound-guided choledochoduodenostomies with fully covered self-expandable metallic stents. World J Gastroenterol
Takagi W, Ogura T, Sano T, et al.
EUS-guided cholecystoduodenostomy for acute cholecystitis with an anti-stent migration and anti-food impaction system; a pilot study. Therap Adv Gastroenterol
Han D, Inamdar S, Lee CW, et al.
Lumen apposing metal stents (LAMSs) for drainage of pancreatic and gallbladder collections: A meta-analysis. J Clin Gastroenterol
Manta R, Mutignani M, Galloro G, et al
. Endoscopic ultrasound-guided gallbladder drainage for acute cholecystitis with a lumen-apposing metal stent: A systematic review of case series. Eur J Gastroenterol Hepatol
Kalva NR, Vanar V, Forcione D, et al
. Efficacy and safety of lumen apposing self-expandable metal stents for EUS guided cholecystostomy: A meta-analysis and systematic review. Can J Gastroenterol Hepatol
Jain D, Bhandari BS, Agrawal N, et al
. Endoscopic ultrasound-guided gallbladder drainage using a lumen-apposing metal stent for acute cholecystitis: A systematic review. Clin Endosc
[Figure 1], [Figure 2], [Figure 3], [Figure 4]
[Table 1], [Table 2], [Table 3]